Mini Surgical Instruments for Veterinary
|FOB Price:||US $420-1,300 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $420-1,300/ Piece|
- Model NO.: BJ8200
- Application: Orthopedic
- Feature: Reusable
- Group: Adult
- Medical Device Regulatory Type: Type 2
- Name: Multi Functional
- Battery Voltage: 7.2V
- Use for: Small Bones ,Foot Hand ,Vets
- Advantage: Lower Noise
- Certificate: Ce,ISO,FDA
- Transport Package: Carton, Aluminun Box
- Origin: Shanghai
- Type: Drill
- Material: Steel
- Certification: CE, SGS, FDA, ISO13485
- Medical Devices Ad. Approval No.: Hxzz20162100264
- Item: Bj8200
- Speed: Variable
- Battery: Rechargeable
- Packing: Carton,Aluminum Box
- Warranty: 2years
- Trademark: bojin
- Specification: 20*50*70
- HS Code: 9018909010
With a comprehensive range of quick change attachments assembled in one full functional handpiece,afford surgeon a convenient option.
Sagittal saw specification:
Bone drill Specification:
Canulate drill Specification:
Attachments for your selection:
1. Sagittal saw.
2. Bone drill.
3. AO bone drill.
4. Canulate drill.
5. Reciprocating saw.
The handpiece guarantees for one year, the battery and other fast consumption items guarantee for 3 months.
1.Our whole power tool can be autoclavable.
2.We can autoclavable for 135 °C (Improved type ≤ 155 °C ),with 1.6Mpa,in this way the bacilli and virus which can not killed by the suffocating will be killed.
3.We use environment friendly nickel-metal hydride battery.
4.The machine has very low noise.
5.Our products has 1 year warranty period.
1.Sterilize temperature ≤ 135 °C (Improved type ≤ 155 °C ),if the temperature is over 135 °C ,the motor will be damaged.
2.Normal bone drill can only hold drill bits less than 6mm,acetabulum reaming drill with big torque,rpm is around 300,canulate drill is used with nail or K wire ,rpm is around 600.
3.The battery need to be get out of the power tool after surgery.Otherwise the small automatic discharge of the electric will make the life of the battery short.
4.The battery cannot be sterilized ,it is not autoclavable.
5.The cranial bur must be washed right after the surgery by purger in order to avoid the blood stick in the bur,then we should make it dry.
6.The battery space in the handpiece of the tool should not have any water,after sterilization.
7.The cranial bur blade is made of medical stainless steel,in order to avoid broken,so it is flexible and not very hard,it is easy to be attrite,we suggest each blade only use for 1 surgery.