Surgical Electric Rechargeable Power Tools Oscillating Saw Sagittal Saw (BJ4101)
|1 2-4 5+|
|FOB Unit Price:||US $2,000 US $1,900 US $1,700|
|Purchase Qty. (Pieces)||FOB Unit Price|
|Production Capacity:||12000 Sets Per Year|
|Transport Package:||Aluminum Case and Carton|
|Payment Terms:||T/T, Western Union, Money Gram|
- Model NO.: BJ4101
- Kind: Orthopedics
- Group: Adult
- Medical Devices Reg./Record No.: Hxzz20162100264
- Power Mode: Battery or AC Voltage
- Color: Silver
- Main Material: Stainless Steel, Aluminium Alloy
- Trademark: Bojin or OEM
- Origin: Shanghai
- Type: Surgical Saw
- Certification: CE, SGS, FDA, ISO13485
- Medical Device Regulatory Type: Type 2
- Applicable Voltage: 220V
- Packing Way: Cartons or Disinfection Box
- Delivery Time: Delivery Within 3 Days After Payment
- Applicate: for Cutting Bone in The Joint Surgery
- Specification: FDA, CE, ISO, SGS
- HS Code: 90189090
Delivery Detail: Within 3 days after payment.
Properties: Orthopedic Surgical Instruments
Type: more powerful sagittal saw
Brand Name: Bojin
Model Number: BJ4101
Place of Origin: Shanghai, China (Mainland)
Main material: Stainless steel, aluminium alloy
Application: joint replacement surgery
For application in joint replacement surgery.
With the perfect shape of ergonomic design,whole unit is autoclavable,motor is imported,low noise, steadily and quickly cutting,no damage for soft tissue,long service life,be warranty for 2 years.
The handpiece guarantees for two years, the battery and other fast consumption items guarantee for 3 months.
1.Our whole power tool can be autoclavable.
We can autoclave for 135 °C (Improved type ≤ 155 °C ),with 1.6Mpa,in this way the bacilli and virus ,for example, AIDS will be killed.
2.We use environmental friendly nickel-metal hydride battery so that battery explosion is impossible.
3.The machine has a very low noise with advanced motor and exact mechanism.
4.Our products have 2 years warranty period and lifetime maintenance.
1.Sterilize temperature ≤ 135 °C (Improved type ≤ 155 °C ),if the temperature is over 135 ° C , the motor will be damaged.
2.Normal bone drill can only hold drill bits less than 6mm,acetabulum reaming drill with big torque,rpm is around 300,canulate drill is used with nail or K wire ,rpm is around 600.
3.The battery needs to be get out of the power tool after surgery.Otherwise the small automatic discharge of the electric will make the life of the battery short.
4.The battery cannot be sterilized , because it is not autoclavable.
5.The cranial bur must be washed right after the surgery by purger in order to avoid the blood stick in the bur,then we should make it dry.
6.The battery space in the handpiece of the tool should not have any water.
7.The cranial bur blade is made of medical stainless steel,in order to avoid breakage,so it is flexible and not very hard,it attrites easily,we advice customers to use each blade for only 1 surgery.